For indirect tissue observation, electron-spin, Overhauser-enhanced, dynamic nuclear polarization magnetic resonance imaging (DNP-MRI) is a useful technique. However, its sensitivity and resolution are low compared with the clinical MRI apparatus. By switching to electron spin resonance (ESR) excitation, the magnetic field of the NMR detec- tion, field cycle technique, which aims to improve resolution, was proposed. However, the eŠect of eddy currents or cur- rent value was altered unsatisfactorily. A team at Kyushu University proposed a new DNP-MRI technique capable of improving NMR detection field by preparing in advance a magnetic field, which was connected by the sample transport system. By developing a mobile MRI method that can be used while moving, and fastening the sample in a disk that rotates at a constant speed, they have developed a circular transport DNP-MRI method that greatly reduces the load on the sample. The circular transport DNP-MRI system comprises a circular sample transport system, detection of an MRI magnetic field of 1.5 T, and ESR excitation magnetic field of 20 mT. The developed DNP-MRI had a clear glass tube phantom and resolution of 0.15 mm, and was successful in imaging multiple radical resonant points. It has been com- mercialized by Japan Redox Limited. In the process of equipment commercialization, a new digital spectrometer has been developed, which expanded the MRI apparatus.
All Science Journal Classification (ASJC) codes
- Pharmaceutical Science